HAMEG 8030 PDF
HAMEG INSTRUMENTS HM | Module: function generator; 7-Segment,5- digit LED; Band: ≤10MHz – This product is available in Transfer Multisort. F requency Range 50 mHz 10 MHz,. Output Voltage up to 10 Vpp (into 50 Ω). ☑ Waveforms: Sine Wave, Triangle, Square Wave, Pulse, DC. ☑ D istortion Factor. Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt, build ?, 4 pictures, 4 schematics, Germany, tubes, semiconductors, Service- or Lab.
|Published (Last):||26 February 2015|
|PDF File Size:||6.37 Mb|
|ePub File Size:||19.81 Mb|
|Price:||Free* [*Free Regsitration Required]|
The selected frequency appears on the 4-digit display 0. The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures required by lead—free soldering. HAMEG will repair or replace products, which prove to be defective during the warranty period.
Safety Every module is manufactured and tested for use only with the mainframe HM according to lEC Part 1 and la Safety requirements for electronic test and measurement equipment. The product detailed below complies with the specifications published by RS Components. hamev
Full text of “Hameg: hm 2”
We regret that transportation damage due to poor packaging is not covered by this warranty. Stability of the output amplitude Setting: With a 50 ohms load at the outputthe signal amplitude should still be Maximum output amplitude Setting: The TP 1 test point is located next to the amplitude control on the upper PC board.
The frequency increases by a factor of The frequency changes by a factor of You agree that the Information as provided here by RS hamef not be error-free, accurate or up-to-date and that it is not advice. General Information Operating condKions: Therefore the maximum output voltage of the output socket is specified with 25Vpp.
The opera- tional position is optional, however, the ventilation holes on the HM and on the plug-in modules must not be obstructed.
To obtain the normal operating temperature, the mainframe with inserted module should be turned on at least 30 minutes before starting the test. Any zeroes preceding the decimal point are dropped. The trig- ger output is short-circuit-proof and can drive several TTL inputs.
If the output of the HM unit comes into contact with components of the circuit under test, which are carrying DC voltage i. With the maximum amplitude of It is TTL compat- ible and has a duty-factor of approx.
The Manufacturers and RS disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for any damages arising from hamev use of or your inability to use the Information downloaded from this website.
No other warranty is expressed or implied. Do not appiy any DC voltage to the out- put hamg. After removal of the two screws at the rear hsmeg the mod- ule, both chassis covers can be lifted.
DC offset If the offset knob is not in its calibrated position, a DC voltage can be superimposed on the output signal. Set the trigger edge con- trol slope of the oscilloscope to negative edge trigger- ing. The set frequency Nq DC voltage not included can be selected at will. Set oscilloscope to DC coupling. We, the Manufacturer or our representatives may use your personal information to contact you to offer support for your design activity and for other related purposes.
Sending feedback, please wait E – Yameg square wave duty cycle Setting: Connect the oscilloscope to output. When removing or replacing the metal case, the instru- ment must be completely disconnected from the mains supply.
Use a 50ohms through-termination. The mains plug of the HM should be inserted before connections are made to measuring circuits. These free programs can be found on this page: Adjust signal height haneg 6 div.
Frequencies are read out on a 4 digit LED display. Electronic Parts List Ref. Pull out the previously used module by its handle and slide in the new module until the end position is reached. In case of any complaint, attach a tag to the instrument with a description of the fault observed. The specified values are only obtained if the amplitude control knob is turned fully clockwise.
A negative DC voltage reduces the frequency.
This voltage is also available in DC mode. Adjust timebase, so that the positive pulse period of the output signal has a width of exactly 10 divisions.